Vicat Softening Temperature ASTM D1525, ISO 306
The Vicat softening temperature is the temperature at which a flat-ended needle penetrates the specimen to the depth of 1 mm under a specific load. The temperature reflects the point of softening to be expected when a material is used in an elevated temperature application.
A test specimen is placed in the testing apparatus so that the penetrating needle rests on its surface at least 1 mm from the edge. A load of 10N or 50N is applied to the specimen. The specimen is then lowered into an oil bath at 23 degrees C. The bath is raised at a rate of 50° or 120° C per hour until the needle penetrates 1 mm.
The test specimen must be between 3 and 6.5 mm thick and at least 10 mm in width and length. No more than three layers may be stacked to achieve minimum thickness.
Temperature at which the needle penetrates 1 mm.
Atlas HDV2 DTUL/ Vicat tester
**Please note that this test description is intentionally generic in nature and aimed at providing a descriptive summary to enhance test understanding. Standards can be obtained from appropriate standard authorities.
Need help or have a question?